**DG4000 series** of generators combines many functions, including Function Generator, Arbitrary Waveform Generator, Pulse Generator, Harmonic Generator, Analog/Digital Modulator and Counter.

**RIGOL DG4162 arbitrary waveform / function generator** adopts the Direct Digital Synthesizer (DDS) technology, can provide stable, precise, pure and low distortion signal. The user-friendly interface design and panel layout of the RIGOL DG4162 gives users exceptional experience. Besides, the remote control of the generator can be easily done through different standard configuration interfaces, which provides more solutions for users.

## RIGOL DG4162 Arbitrary Waveform / Function Generator User Manuals

Download RIGOL DG4162 arbitrary waveform / function generator user manual

## RIGOL DG4162 Arbitrary Waveform / Function Generator Software

Download RIGOL DG4162 arbitrary waveform / function generator software

## RIGOL DG4162 Arbitrary Waveform / Function Generator Features

- 6 standard waveforms:
- Sine
- Square
- Ramp
- Pulse
- Noise
- Harmonics

- Arbitrary waveform function with up to 150 built-in waveforms
- 500 MSa/s sample rate
- 2 ppm high frequency stability, -115 dBc/Hz low phase noise
- Build-in high precision 200 MHz BW frequency counter
- Up to 16 orders Harmonic generation function
- 7 inch color LCD (800 × 480 pixels)
- 14 bits vertical resolution
- 16 K waveform length

## RIGOL DG4162 Arbitrary Waveform / Function Generator Technical Specifications

Standard waveforms | ||||
---|---|---|---|---|

Waveforms |
Sine, Square, Ramp, Pulse, Noise, Harmonics Arbitrary waveforms |
|||

Sine | 1 μHz to 160 MHz | |||

Square | 1 μHz to 50 MHz | |||

Pulse | 1 μHz to 40 MHz | |||

Ramp | 1 μHz to 4 MHz | |||

Harmonic | 1 μHz to 80 MHz | |||

White noise | 120 MHz (-3 db) | |||

Resolution | 1 μHz | |||

Sine wave spectral purity | ||||

Harmonic distortion | DC-1 MHz | 1 MHz-10 MHz | 10 MHz-100 MHz | 100 MHz-160 MHz |

<-60 dBc | <-55 dBc | <-50 dBc | <-40 dBc | |

Total harmonic distorion | <0.1%(10 Hz-20 kHz, 0 dBm) | |||

Spurious (non-harmonic) | ≤10 MHz | <-65 dBc | ||

>10 MHz | <-65 dBc + 6 dB/octave | |||

Phase noise | 10 MHz: ≤-115 dBc/Hz | |||

Square wave characteristic | ||||

Rise/fall time | <8 ns | |||

Overshoot | <3% | |||

Duty cycle | ≤10 MHz | 20.0% to 80.0% | ||

10 MHz-40 MHz | 40.0% to 60.0% | |||

>40 MHz | 50.0% （fixed） | |||

Jitter | ≤5 MHz 2 ppm + 500 ps | |||

Pulse wave characteristics | ||||

Pulse period | 25 ns to 1000000 s | |||

Pulse width | ≥10 ns | |||

Overshoot | <3% | |||

Jitter | ≤5 MHz 2 ppm + 500 ps | |||

Arbitrary wave characteristics | ||||

Waveform length | 16 K points | |||

Vertical resolution | 14 bits | |||

Sample rate | 500 MSa/s | |||

Minimum rising /falling time | <5 ns | |||

Jitter (RMS) | ≤5 MHZ 2 ppm + 500 ps | |||

Interpolation method | Close, Linear | |||

Output characteristics | ||||

Amplitude | ≤20 MHz： 1 mVpp to 10 Vpp ≤60 MHz： 1 mVpp to 5 Vpp ≤120 MHz： 1 mVpp to 2.5 Vpp ≤160 MHz： 1 mVpp to 1 Vpp |
|||

Amplitude accuracy (1 kHz) |
± 1% of setting + 2 mVpp | |||

Amplitude flatness (sine wave relative to 100 kHz) | ≤10 MHz： ±0.1 dB ≤60 MHz： ±0.2 dB ≤100 MHz： ±0.4 dB ≤160 MHz： ±0.8 dB |
|||

DC offset | ||||

Range | ±5 Vpk ac + dc | |||

Accuracy | 1% of setting + 5mV + 0.5% of amplitude | |||

AM modulation | ||||

Carrier waveforms | Sine, Square, Ramp, Noise, Arb | |||

Source | Internal / external | |||

Modulating waveforms | Sine, Square, Ramp, Noise, Arb | |||

Depth | 0% to 120% | |||

FM modulation | ||||

Carrier waveforms | Sine, Square, Ramp, Arb | |||

Source | Internal / external | |||

Modulating waveforms | Sine, Square, Ramp, Noise, Arb | |||

Modulating frequency | 2 mHz ～ 50 kHz | |||

PM modulation | ||||

Carrier waveforms | Sine, Square, Ramp, Arb | |||

Source | Internal / external | |||

Modulating waveforms | Sine, Square, Ramp, Noise, Arb | |||

Phase deviation | 0° to 360° | |||

ASK modulation | ||||

Carrier waveforms | Sine, Square, Ramp, Arb | |||

Source | Internal / external | |||

Modulating waveforms | 50% duty cycle square | |||

FSK modulation | ||||

Carrier waveforms | Sine, Square, Ramp, Arb | |||

Source | Internal / external | |||

Modulating waveforms | 50% duty cycle square | |||

3FSK modulation | ||||

Carrier waveforms | Sine, Square, Ramp, Arb | |||

Source | Internal | |||

Modulating waveforms | 50% duty cycle square | |||

4FSK modulation | ||||

Carrier waveforms | Sine, Square, Ramp, Arb | |||

Source | Internal | |||

Modulating waveforms | 50% duty cycle square | |||

PSK modulation | ||||

Carrier waveforms | Sine, Square, Ramp, Arb | |||

Source | Internal / external | |||

Modulating waveforms | 50% duty cycle square | |||

BPSK modulation | ||||

Carrier waveforms | Sine, Square, Ramp, Arb | |||

Source | Internal | |||

Modulating waveforms | 50% duty cycle square | |||

QPSK modulation | ||||

Carrier waveforms | Sine, Square, Ramp, Arb | |||

Source | Internal | |||

Modulating waveforms | 50% duty cycle square | |||

OSK modulation | ||||

Carrier waveforms | Sine | |||

Source | Internal / external | |||

Oscillation time | 8 ns ～ 200 s | |||

PWM modulation | ||||

Carrier waveforms | Pulse | |||

Source | Internal / external | |||

Modulating waveforms | Sine, Square, Ramp, Noise, Arb | |||

Width deviation | 0% to 100% of Pulse Width | |||

Sweep | ||||

Carrier waveforms | Sine, Square, Ramp, Arb | |||

Type | Linear, log or step | |||

Direction | up or down | |||

Start/Stop Frequency | 1 μHz to 160 MHz | |||

Sweep time | 1 ms to 300 s | |||

Source | Internal/ external/ manual | |||

Burst | ||||

Waveforms | Sine, Square, Ramp, Pulse, Noise, Arb | |||

Carrier Frequency | 2 mHz to 100 MHz | |||

Burst Count | 1 to 1 000 000 or Infinite | |||

Internal period | 2 μs to 500 s | |||

Gate source | External trigger | |||

Trigger source | Internal/ external/ manual | |||

General specification | ||||

Display | 7-inch TFT LCD (800 × 480 pixels) | |||

Interfaces | USB Host (2), USB Device, LAN | |||

Power | 100 V ～ 240 V (45 Hz ～ 440 Hz) Less than 50 W |
|||

Dimentions | 313 × 160.7 × 116.74 mm | |||

Weight | 4.5 kg |

## RIGOL DG4162 Arbitrary Waveform / Function Generator Package Contents

- RIGOL DG4162 arbitrary waveform / function generator (1 pc.)
- Power cord (1 pc.)
- USB cable (1 pc.)
- BNC cable, 1 meter (1 pc.)

## RIGOL DG4162 Arbitrary Waveform / Function Generator Optional Accessories

- 40 dB Attenuator (1 pc.)
- Rack Mount Kit (1 pc.)